|Place of Origin:||China|
|Certification:||KEMA CE ISO SGS TUV UC3|
|Minimum Order Quantity:||1pcs|
|Delivery Time:||3-8 days|
|Payment Terms:||L/C, TT,D/A,D/P|
|Material:||Epoxy Resin||Frequency Range:||10 Hz To 20 KHz (-3dB)|
|Installation Site:||Indoor||Range Of Current Measurement:||0.5 - 5000 A|
|Linearity:||±0.3% (5%~120%In)||Standard:||GB 20840,GB 4208-2008,IEC/EN 61869,IEEE C57.133.3,IEC 60529|
|Accuracy:||±1%(5%~120%In) @45-65Hz||Output:||0 - 5 A（RMS）|
|Variable Range Stalls:||500A，1kA，2.5kA，5kA||Max Burden:||1.25 VA Each Channel|
|Max Output Impedance:||50 MΩ||Frequency:||50/60 Hz|
flexible current transformer,
High Electrical Efficiency CY-RCT02A1(5) Clamp-on Flexible Rogowski Coil Split Current Transformer
"CY-RCT02A1 (5)" Type 1 (5)A Rogowski coil split current transformer provides easy-to-install plug-and-play solution that can be connected to a power meter or a 1(5)A current transformer Equipment used in conjunction.
The Rogowski coil split current transformer Kit has a three-channel 1(5)A integrator output that can be used for most power meters and devices with 1 (5) A input. The kit offers four selectable measurement ranges including: 500A, 1kA, 2.5kA and 5kA, the detection range of 0.5% to 120% of the rated current input .Applied to 50 / 60Hz system, the rated frequency range of 45Hz to 65Hz. The mounting bracket is DIN rail or surface mount, with 24Vdc power supply.
Application :Uses: power instrumentation, electronic multi-function energy meter and field calibrator, instrumentation measurement and protection.
No need to disconnect the bus line power outage, easy installation.
Simple and quick installation in a small space;
Fine design, coil size can be customized
Remarks: Upon request we are glad to offer transformers according to other standards or with non-standard technical specs..
When flexible rogowski coil measurement probe connected, the cable under test should be placed in the middle of the flexible rogowski coil and the coil can be kept vertical to effectively reduce the measurement error.
in daily use , please ensure the correct connection of the XS1 terminal＇s positive and negative polarity of the power supply, once incorrect connection, the inside chip will be damaged.
If the output channel for more than 5 amps RMS may cause RōCoilTCA-5 to overheat and damage.
Contact Person: Sales Manager